I am attaching a vi which i tried. Then simply wire an empty array to initialize the two Plot Array shift registers which will start the plots over again. I have two modes of experiment. Email Required, but never shown. What are you using for a temperature sensor? However, since moving over to Windows 10, I can’t find where to install those onto my machine, to the point where I am That’s just index array, which gives you the 1-d array that’s relevant.
|Date Added:||11 September 2017|
|File Size:||67.50 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
I did not know that in thermoluminescence the data collection should be continuous.
After starting the experiment, at every 5 or 10 Labvjew increment in temperature again to be set by the usera signal should be sent to the trigger to record the spectrograph laoeshore the temperature rise should not stop. Message 4 of 9.
When you really only want to plot data of the current sweep you need to delete the data stored in the shift registers aka feedback nodes when you start a new sweep!
Newest ‘visa’ Questions – Stack Overflow
Accepted by topic author RSGR. Check the controller failure status etc. Violet 37 1 5. If you want to do something similar in the first loop, which uses an Express VI boooooinvestigate what the Reset terminal does my lakesuore is that it “resets”, or zeros, the plot. Without having a temperature controller to try this out on, there is no way for us to know how the controller responds to these commands. I am attaching a vi which i tried.
Ensure that comms exist between the software and the controller etc. Thanks for the reply. If you want to clear the history you can send an lsbview array to history property using a property node.
Lake Shore Cryotronics 330 Controller
Model | Downloads | Lake Shore Cryotronics, Inc.
Let me explain the experiment. Also for timing purposes, search for functional global timer or try modify this example https: Sure, you have just have to be careful about how you send the data to the graph. I am stepping the temperature and sweeping voltage. Message 9 of I am not sure if the spectrograph I have works continuously or not. If you are trying to use the HIGH output for heater power to achieve fine control at low temperatures, you’re going to have a bad labvuew.
I need some help again. My source is very simple: The Keithley is connected to the computer via a a RSto Esben Jensen 11 1 6.
Anyway it works now, thank you for the aid. It would be very helpful if anyone can guide me in the right way. I tested it manually and it is working fine.